Rainfall this month was only 53.3% of the 30 year
average for July in Bristol. Between the 26th June and July 16th there was
only 0.2 mm of rain recorded.
It was the 5th month of 2019 where rainfall was below average.
temperature for the month of 19.6°C was 1.3°C above the 30 year average for
Bristol. The high of 33.2°C on the 23rd was the Highest July temperature
since 34.1°C on July 19th 2019 and was the 4th warmest July on record since
this data started in 1937. The highest July temperature in Bristol was
35.4°C recorded also on the 19th was in 2006.
It was a relatively
calm month with below the average wind speed for July. There were a
predominance of south westerly winds with 18 days coming from that
2018 Annual summary is here:
For the full monthly summary please go here:
Meteorological site is situated in an urban housing area approximately one
mile to the south of the Centre of Bristol.
This area is in a district called Totterdown.
Statistical correlations with the Bristol
Meteorological Office site, that was
less than two miles to the North and was
situated on top of a City Centre office block, were generally good with
rainfall and barometric data particularly close.
The site of the station is approximately 34 metres above sea level with the
manual instruments mounted in a Stevenson screen. The screen is sited
centrally in a small concreted garden area. A Snowdon
rain gauge is mounted correctly in the ground but does not have the full open
space required around it. A Davis
Vantage Pro2 Plus automatic weather station (AWS) was
added in May 2005 and this is mounted 4 feet above a concrete shelter. The
wind data is gathered automatically from a Davis Vantage Pro2 Plus anemometer which is
sited 3.3 metres (10 feet) above the house top.
Manual observations are still taken at approximately 09:00 hours GMT every
day for rainfall and cloud cover. Rainfall manually recorded in the Snowdon rain gauge is entered for the previous day. The
wind data is constantly logged and automatically reset at 24:00 hours GMT
every day (in the summer BST -1 hour. The Davis rainfall tipping bucket was recalibrated in November 2005 and is now
consistently under 3% under recording (an improvement from installation in
May 2005 when 10% was more
typically the norm).
Since the arrival of the Davis
Vantage Pro2 Plus equipment in May 2005 temperatures, humidity and barometric
pressure are recorded from the new equipment. These figures will be more
accurate and will be taken at exactly 09:00 GMT (in the summer BST -1 hour)
as the data is now logged to a computer.
By default the Davis
equipment records every parameter for the 24 hour period 00:00 to 23:59 BST
or GMT. Data analysis will enable manual data extraction of temperatures and
rainfall to 09:00 GMT (adjusted in the summer). Due to the 0.2 mm (tipping
bucket) measurements of precipitation recorded by the Davis equipment the
Snowdon rain gauge will still be used due to its greater accuracy (see